Text preview for : TDSHT3_Quick_Ref_Card_SourceTest.pdf part of Tektronix TDSHT3_Quick_Ref_Card_SourceTest Tektronix TDS7000 TekScope_Opt_App_1.0.10_071-1883-01_063-3914-10 TekScope_Opt_App_1.0.10_071-1883-01_063-3914-10 Documents HT3-1.2a_Documentation TDSHT3_Quick_Ref_Card_SourceTest.pdf
Back to : TDSHT3_Quick_Ref_Card_Sou | HomeGetting Started with TDSHT3 4. To connect the DUT, click Tests > Connect. 8. If you have run the tests successfully, the software makes Result available automatically and displays theYou can use this side of the quick reference card to get eye diagram plot and the clock jitter plot. You can alsostarted with the Source Clock-Data Tests Select All test view both the result summary of the test and the reportprocedure. The other side contains a complete menu tree for configuration in the result pane.the TDSHT3 HDMI Compliance Test Software. NOTE. For additional procedures, refer to the TDSHT3 Quick Start User Manual. For complete operating instructions, refer to the online help. 5. Make the connections as follows:TDSHT3 HDMI Compliance Test Software enablesunprecendented efficiency by providing a comprehensiverange of tests, including Jitter Tolerance. 9. In the result summary pane, click Result Details toSource Clock-Data Tests Select All display the results of the tests.This option enables you to run the Eye Diagram, DutyCycle, Rise Time, Fall Time, Clock Jitter, Over/UndershootV-H, Over/Undershoot V-I, and Inter-Pair Skew testsimultaneously.You will need one supported oscilloscope, two differentialprobes, one DC power supply 3.3 V, one EDID emulator,and one TPA-P-DI fixture.Follow these steps:1. On the menu bar, click Tests > Select > Source. 10. In the Result Details dialog box, click Result Statistics2. In the clock-data tests pane, click Select All. to display statistics based on the tests. 6. Ensure that your signal in the oscilloscope display is similar to the sample signal. Click View Waveform to display a sample of the expected signal. If the displays are not similar, go back and check your configuration and connections.3. To change the configuration settings, click Tests > Configure. For most tests, you can use the factory The software calculates statistics for each selected test, and default configuration. However, you can change the logs the statistics on a cycle-by-cycle basis in a fairly large values by using the virtual keyboard or the general waveform. The standard statistics are for the Maximum, purpose knob on the oscilloscope front panel. Using the Minimum, Mean, Standard Deviation, and Population. File menu, you can also restore the factory defaults or save and recall your own configuration settings. 7. Click Run Test to perform the test. The TDSHT3 HDMI Compliance Test Software sets up the oscilloscope and the test runs displaying a progress indicator.For up-to-date information on Tektronix oscilloscope To order along with oscilloscope:solutions for HDMI Compliance Test Software, access thewww.tektronix.com Web page. H Opt. HT3 - HDMI Compliance Test Software To order an upgrade for an existing oscilloscope:TDSHT3 Ordering Information H TDS6000B - Order TDS6BUP - Opt. HT3This software supports the TDS6000B, TDS6000C,TDS7254/B, TDS7404/B, CSA7404/B, TDS7704B, and H TDS7000/B - Order TDS7UP/7BUP - Opt. HT3DPO/DSA70000 series of oscilloscopes. Refer to the H CSA7000/B - Order TDS7UP/7BUP - Opt. HT3Optional Applications Software on Windows-BasedOscilloscopes Installation Manual for a list of specific Recommended Accessoriesmodels. The applications CD includes a PDF file of theonline help. Oscilloscopes H 16M Record Length/Ch - Opt. 4M or more (for Eye Diagram and Jitter tests) H Serial Trigger - Opt. ST (for Inter-pair Skew tests) HDMI Pattern Sources H Jitter Generation - AWG710B (for Sink tests) H Test Pattern Generation - DTG5274 and DTGM30 for Sink Tests (three required) TDR Tests H Oscilloscope - TDS8000B with 80E03 and 80E04 modules TDSHT3 Probes HDMI Compliance Test Software H Differential Probes - P6330, P7330, P7350 (two probes Reference required), and P7380 H Active Probes - P7240 and above (two probes required) Source Test H Probe Positioner - PPM100 Flexible Arm Probe Posi- tioner Test Fixtures H Plug type - HDMI TPA-P adapter set - 013-A013-50 H Receptacle type - HDMI TPA-R adapter set - 013-A012-50 www.tektronix.com *P071195901* 071-1959-01TDSHT3 Menu Tree Recall Default Recall Save File Recall Recent Position Eye Mask in Center Preferences Acquisition Alert Message Minimize Pulse Width Trigger (Rise/Fall Time or Over/Undershoot) Exit Edge Trigger Eye Diagram Source Duty Cycle Select Sink Rise Time Cable Fall Time Clock Jitter Over/Undershoot VH Source Over/Undershoot VL InterPair Skew (ClockData Tests) InterPair Skew (DataData Tests) IntraPair Skew Low Amplitude + Tests Configure Low Amplitude Min/MaxDiff Swing Tolerance Sink Jitter Tolerance IntraPair Skew Cable Eye Diagram Connect View Waveform Summary Results Details Define Test Template Utilities Report Generator Define Report Layout Deskew Generate Report Help Help Topics About TDSHT3Copyright Tektronix, Inc.