Text preview for : www.thinksrs.com-SR620_Hyp_Term.pdf part of Stanford Research Systems www.thinksrs.com-SR620 Hyp Term Stanford Research Systems www.thinksrs.com-SR620_Hyp_Term.pdf
Back to : www.thinksrs.com-SR620_Hy | Home
Using HyperTerminal to communicate with an SR620
Make sure that your connected to the Serial Port and not the Printer Port on the rear panel
1. Power on reset the SR620
2. Open Hyper Terminal and name your connection. Then Click OK.
3. Make sure the COM port is correct. Then Click OK.
4. Set the Bits per second: to 9600, Data bits: 8, Parity: None, Stop bits: 1, Flow control: Hardware.
5. Go to the Properties Menu under the File Tab.
6. Click on Settings then ASCII Setup.
7. Put a check mark in the box next to Echo typed characters locally, then Click OK and then
OK to exit the properties.
8. Hit the return key several times. The "ACT" light on the SR620 should turn on and off.
9. Type *idn? , Then the return key
10. Hyper Terminal should return the line "StanfordResearchSystems,SR620,0xxxx,1.48" (xxxx=the serial
number of the instrument"