|Fluke_803D_Army.pdf||TM 9-4935-282-34 DEPARTMENT OF THE ARMY TECHNICAL MANUAL
FIELD MAINTENANCE MANUAL
VOLTMETER, ELECTRONIC JOHN FLUKE MODEL 803D FSN 6625-072-4303
This copy is a reprint which includes current pages from Change 1.
HEADQUARTERS, DEPARTMENT OF THE ARMY MARCH 1964
TM 9-4935-282-34 C1
RADIATION HAZARD This equipment contains the following radioactive items: Nomenclature OA2 Vacuum tube OA2 Vacuum tube OA2 Vacuum tube OG3-2 Vacuum tube FSN 5960-503-4880 5960-503-4880 5960-503-4880 5960-912-7630 Isotope UO 2 Ni63 Co60 Unknown Amount (Microcuries) 0.10uc 0.50uc 0.20uc Unknown
Refer to TM 3-261, TM 38-250, and TB 750-237 for information relative to shipping, storage, handling, and disposal of radioactive material. FIRST AID FOR RADIOACTIVE CONTACT The following first aid procedure for wounds caused by anything coated with a radioactive material represents the only reasonable first aid treatment which would possibly be available. a. Stimulation of mild bleeding by normal pressure ab|