|Making Reflection Measurements - Application Note 5991-2082EN c20140726 .pdf||Keysight Technologies
Making Reflection Measurements
Using the N9322C
Basic Spectrum Analyzer (BSA)
In addition to measuring the frequency power characteristics of your
device or system with a spectrum analyzer, sometimes you need to
evaluate the performance characteristics of antenna, RFID tags,
or RF Tx modules, such as their return loss, insertion loss, and VSWR.
The Keysight Technologies, Inc. N9322C basic spectrum analyzer (BSA)
supports those additional measurements through the use of an optional
tracking generator (Option TG7) and reflection measurement (Option
RM7). The built-in bridge in the tracking generator makes the N9322C
an easy-to-use reflection analyzer.
Introduction to Reflection Measurement
A scalar reflection measurement|