Now downloading free:LeCroy LECROY AP034-OM-E-06 Proformace
Download LeCroy LECROY AP034-OM-E-06 Proformace docs - Various electronics service manuals
>> View document online <<
>> Download document <<
|Information about the files in archive:|
|File name||Text |
|LECROY AP034-OM-E-06 Proformace.pdf||3HUIRUPDQFH 9HULILFDWLRQ
This procedure can be used to verify the warranted
characteristics of the AP034 Active Differential Probe.
The recommended calibration interval for the model AP034
Active Differential Probe is one year. The complete performance
verification procedure should be performed as the first step of
annual calibration. You can record test results on a photocopy of
the Test Record provided at the end of this section.
You can do the performance verification without removing the
instrument covers, exposing yourself to hazardous voltages.
Adjustment should only be attempted if a parameter measured in
the Performance Verification Procedure is outside of the
Average rating for this file: 0.00 ( from 0 votes)Similar Service Manuals :
- - - - - - -