|English _ 2012-10-02 _ PDF 381 KB 5991-1130EN c20121018 .pdf||Agilent 33500 Series
The Agilent 33522A dual-channel waveform generator To use these features from the instrument's front panel,
allows you to output waveforms on both channels indepen- press the channel output key above either channel's
dently. This allows you to flexibly drive a device under test output connector. Then press More > Dual Channel on
(DUT) in many common situations. However, some applica- the softkey menu. The menu shown in Figure 1 appears,
tions require you to combine two signals on one channel; showing the current state of dual-channel features.
for example, you may wish to add noise to a sine wave. Dual-channel operations are disabled in the factory
Other applications may require signals that are related to default state.