Now downloading free:Fluke aug96pp2
Download Fluke aug96pp2 docs - Various electronics service manuals
|File name:||aug96pp2.pdf |
[preview aug96pp2] new window
|Descr:|| Fluke appnotes aug96pp2.pdf|
|Group:|| > > |
|Information about the files in archive:|
|File name||Text |
|aug96pp2.pdf|| MAINTAINING 10 VDC AT 0.3 PPM OR BETTER
IN YOUR LABORATORY
There is a need today to maintain dc voltage at an uncertainty of 0.3 ppm or better. We
believe that most standards laboratories can do this without using an expensive Josephson
Junction Array System. DC Reference Standards commonly in use today are capable of
this performance providing that they are properly calibrated and that the effects of
temperature, pressure and seasons are taken into account. This paper develops supporting
uncertainty equations and applies them to useful scenarios.
High end multifunction calibrators and digital voltmeters today require calibration
uncertainties that only primary standards laboratories maintained in recent history.
Uncertainties of 1.0 to 1.5|
|Download restrictions information:|
|Month||5000||7||Wrong count ?|
Download >> To download the file, please, click here !
Did You find this Service Manual useful? Share it:
Was this file useful ? Share Your thoughts with the other users.
User ratings and reviews for this file:
Average rating for this file: 0.00 ( from 0 votes)Similar Service Manuals : - - - - - - -