|U1253Bfirmware_release_2.26 Revision history notes for U1253B firmware.txt||Firmware revision updates:-
Version 2.26 fixed an issue present in the previous release.
- Fix incorrect reading when the battery level is at = 4.7V. Wrong reading on the display may mislead users about the measurement made.
- Fix datalogging memory retrieval problem. When log memory is full, it will not be able to retrieve previously logged data.
- Fix inaccurate DC measurement when mixed signal (AC+DC) present. The factory default option is set to DC Filter ON during DC measurement. The DC Filter function attenuates AC signals to help you read the correct DC measurement from a mixed signal.
- Initial release|