|model_2188_low_thermal_calibration_shorting_plug.pdf|| Model 2188 Low-Thermal Calibration Shorting Plug
The Model 2188 is shown in Figure 1. This low-resistance, low-thermal shorting plug is required to calibrate the
Model 2182 Nanovoltmeter. As shown in Figure 2, all four terminal pins are shorted together.
The shorting plug is made up of a LEMO connector whose pin connectors are shorted together using copper wire
and SN/AG4 silver solder.
Input Terminal Cleaning
Before connecting the shorting plug to the LEMO input connector on the Model 2182, clean the copper contacts of
the connectors using DeoxIT D100L deoxidizing solution as follows:
NOTE The following procedure uses DeoxIT to remove oxidation from the contacts of the shorting plug and the LEMO
input connector. A bottle of DeoxIT is supplied with the Model 2182 Nanovoltmeter.
1. Turn off the Model 2182 and, at the rear panel, disconnect the line cord and any other cables or wires connected
to the instrument.
2. Stand the Model |