Download Fluke 4225366C w docs - Various electronics service manuals
|File name:||4225366C_w.pdf |
[preview 4225366C w] new window
|Original Chassis:||4225366C w|
|Descr:|| Fluke 5502A 4225366C_w.pdf|
|Group:|| > > |
|Information about the files in archive:|
|File name||Text |
The following tables list the 5502A specifications. All specifications are valid after allowing a warm-up period of
30 minutes, or twice the time the 5502A has been turned off. (For example, if the 5502A has been turned off for
5 minutes, the warm-up period is 10 minutes.)
All specifications apply for the temperature and time period indicated. For temperatures outside of tcal 5 C (tcal is the
ambient temperature when the 5502A was calibrated), the temperature coefficient as stated in the General Specifications
must be applied.
The specifications also assume the Calibrator is zeroed every seven days or whenever the ambient temperature changes
more than 5 C. The tightest ohms specifications are maintained with a zero cal every 12 hours within 1 C of use.
Also see additional specifications later in this chapter for information on extended specifications for ac voltage and
Warmup Time ...|
|Download restrictions information:|
|Month||5000||22||Wrong count ?|
Download >> To download the file, please, click here !
Did You find this Service Manual useful? Share it:
Was this file useful ? Share Your thoughts with the other users.
User ratings and reviews for this file:
Average rating for this file: 0.00 ( from 0 votes)Similar Service Manuals : - - - - - - -