Now downloading free:Keithley 2430SweepPulses
Download Keithley 2430SweepPulses docs - Various electronics service manuals
|File name:||2430SweepPulses.pdf |
[preview 2430SweepPulses] new window
|Descr:|| Keithley 2430 2430SweepPulses.pdf|
|Group:|| > > |
|Information about the files in archive:|
|File name||Text |
|2430SweepPulses.pdf||Is it Possible to Combine Sweep Operation with Pulse Mode on the Model 2430?
It is very much possible to do a Sweep of the Pulses with the Model 2430. It can be
accomplished from either the Front Panel or via bus control (GPIB or RS-232). First, define the
Pulse characteristics (pulse width, pulse delay). Having done this, set up a Sweep by selecting
the Sweep type (linear, log, custom) and configuring the sweep points.
As an example, the scope trace below shows a linear pulse sweep. The Sweep starts at 1 V and
ends at 10 V, with a step size of 1 V. This was set for an Infinite sweep count. In the trigger Model
set the Arm Count for Infinite and the Trigger count for 10.
If selecting the Log sweep type, make sure you do not start the sweep from 0 Volts or 0 Amps as
the logarithm of zero cannot be calculated.
For more information, see section 5 (Pulse Mode) and section 10 (Sweep Operation) in the Model
2400 Series SourceMeter User's Manual.
|Download restrictions information:|
|Month||5000||5||Wrong count ?|
Download >> To download the file, please, click here !
Did You find this Service Manual useful? Share it:
Was this file useful ? Share Your thoughts with the other users.
User ratings and reviews for this file:
Average rating for this file: 0.00 ( from 0 votes)Similar Service Manuals : - - - - - - -