|KPCI-3108_EXP16.PDF||Using EXP-16 or EXP-16/A with KPCI-3108 Series
When an application calls for external multiplexing, the KPCI-3108 series of board should be used with the
EXP-1800 external multiplexer for best results. The EXP-1800 has very fast settling time capabilities and
can switch at the full speed rating of the KPCI-3108.
However, the KPCI-3108 does provide for upgrades from the DAS-1600 series systems. For those with
significant investment in signal conditioning designed for use with DAS-1600 (EXP-GP, EXP-16, etc), an
easy connection is provided to the KPCI-3108. Operational and performance differences need to be taken
into account in the system implementation.
The STA-3108-A2 screw terminal panel for use with the KPCI-3108 Series provides a DAS-1600 Series
compatible male DB37 connector. For example, to connect to an EXP-16 or EXP-16/A see figure below:
Expansion Channel Addressing and Configuration
Expansion channel addressing should be enabled in the DriverLINX co|