Now downloading free:Keithley 4440 RTD Require
Download Keithley 4440 RTD Require docs - Various electronics service manuals
|File name:||4440_RTD_Require.pdf |
[preview 4440 RTD Require] new window
|Original Chassis:||4440 RTD Require|
|Descr:|| Keithley Appnotes 4440_RTD_Require.pdf|
|Group:|| > > |
|Information about the files in archive:|
|File name||Text |
|4440_RTD_Require.pdf|| RTD Instrumentation Requirements
By Tom Hayden and Joel Roop
Keithley Instruments, Inc.
Temperature measurement is vital in many processes, during product
development, in safety systems and a host of other applications. For the majority
of these applications, a temperature sensor in contact with a solid surface, or
immersed in a fluid, can be used. The most commonly used contact sensors are
thermistors, thermocouples, and RTDs (resistance temperature detectors).
An RTD frequently is the sensor of choice when sensitivity and application
flexibility are the most important application criteria. They can be manufactured in
a variety of configurations and used for measurements covering a wide
temperature range. When a 100ohms RTD is properly excited (~1mA), it will
produce a 400 |
|Download restrictions information:|
|Month||5000||45||Wrong count ?|
Download >> To download the file, please, click here !
Did You find this Service Manual useful? Share it:
Was this file useful ? Share Your thoughts with the other users.
User ratings and reviews for this file:
Average rating for this file: 0.00 ( from 0 votes)Similar Service Manuals : - - - - - - -