|Wavetek_3510_Signal_Generator_Service_Manual_(partial).pdf|| Wavetek 3510 Service Manual
1.1 CALIBRATION PROCEDURE
Remove the instrument top and bottom cover and the card cage cover. Allow a
two-hour warm-up period before calibration. In general, calibration should be
performed in the sequence given.
1.1.1 Power Supply
Note: All measurements are made with reference to chassis ground.
+18V VOLT ADJUSTMENT |